Implant And Abutment
Level Impressions Twist Lock™ Transfer Impression
Select the proper Twist Lock Impression Coping
by matching the EP« Diameter of the healing abutment and
matching the color of the implant platform to determine
platform diameter (see below). Remove the healing abutment
from the implant using the Large Hex Driver. To help prevent
accidental swallowing, thread floss through the spinner on the
Activate the fingers using the QuickSeat« Activator Tool.
Place the Twist Lock Impression Coping into the implant, line
up the hex and press firmly until feeling the tactile click.
Place the Twist Lock Impression Coping on the
implant and engage the hex.
Thread the Twist Lock Impression Coping Screw
into the implant until finger tight. Tighten the screw using
the Impression Coping Driver.
3. Radiograph the interface to verify complete seating
of the coping on the implant. Place the film perpendicular to
the interface of the coping on the implant or abutment.
4. A custom or stock closed top impression tray is used
for the Twist Lock Transfer Impression technique. Try in the
tray to verify that there is no contact with the coping.